- About Valeport
- Our international distributors
- Services and Calibrations
- Sound Velocity Sensors
- Support Form Response
- Technology you can trust
- Terms & Conditions
- Warranties and Repairs
Your browser is no longer supported. Please upgrade your browser to improve your experience.
In this video we will be demonstrating the maintenance, care and calibration checks of the Dissolved Oxygen (DO) sensor of the Midas CTD+ instrument.
Instruction video showing how to check the calibration of pH Sensors.
Instruction video showing CTD+ pH and REDOX sensor maintenance and care.
See how to replace the lithium battery on the MIDAS SVX2 Combined CTD/ Sound Velocity Profiler.
See how to replace the batteries and O rings for your Valeport Midas SVX2 / SVP / CTD instruments. Visit http://www.valeport.co.uk for more info.
Step by step guide as to how to change the Impeller and Shaft on the Model 106 Current Meter.
See how to service your Valeport 106 Current Meter. More info at http://www.valeport.co.uk
Step by step guide as to how to change the battery in the Valeport Model 740 Tide Guage.
Learn how to change the C Cell Battery in the Tidemaster Portable Tide Gauge from Valeport. Visit http://www.valeport.co.uk/Support for more info.
Instructions on how to change pressure to radar on the Valeport Tidemaster Tide Gauge. Visit http://www.valeport.co.uk/Support for more info
Support Video showing how to install a new Winstar OLED display on the Valeport Tidemaster.
SubConn® greasing and mating instructions. Demonstration video showing how to grease and mate SubConn® connectors above water (dry mate)
SubConn® greasing and mating instructions. A demonstration video showing how to grease and mate SubConn® connectors underwater (wet mate)
SubConn® cleaning instructions. A demonstration video showing how to clean the female half of SubConn® connectors.